Precautions for CE Markings Keyence confirms that the LS Series . LS- Photoelectric Sensor or Proximity Sensor Timing input signals are sent. Find great deals for 1 PC KEYENCE Ls CCD Laser Micrometer. Shop with confidence on eBay!. LS High-performance controller. Visible measurement pointController with Target Viewer. High-durability CCD . The LS Series measures the outer diameter of a fiber continuously and .. E-mail: [email protected]
|Published (Last):||24 July 2018|
|PDF File Size:||4.31 Mb|
|ePub File Size:||17.47 Mb|
|Price:||Free* [*Free Regsitration Required]|
The result output of a difference check is a Chapter 4 Operation Control Note: Press the [Up] key and move the cursor to underline the type of calibration. Be the first to review this item. Setting Range Set the following items.
Setting Range The range is initially set to — Status display Target viewer display Displays the image of the measuring target. Please read the manual carefully and be sure you understand the information provided before attempting to install and operate the LS Series. Appendix Index 1 One-point calibration Chapter 5 Function Settings 4 Press the [Up or Down] key and move the cursor frame to the area to be set.
Types of Limits The following tolerance settings are possible. Withoutabox Submit to Film Festivals. If the measured value exceeds the range of the Y-axis of the graph, the value will be displayed as the maximum or minimum value in the Y-axis direction.
The image of the target does not appear. A change in the ambient operating temperature may result in measurement errors. Page 50 Chapter 3 Easy Setting Guide Statistical Processing In the following example, the values calculated from the measuring data are: Page Appendix Environment Setting Record Sheet Use the following tables to record your environment setting details.
The types of measured values held vary with the MEA-Mode. Furthermore, group values can be statistically processed.
These settings can be registered as programs that can be used whenever required. Set the number of measuring times.
Page Chapter 6 Environment Settings Details of Environment Settings This section provides information on environment setting items and setting methods. The following items are displayed for setting. Page 11 Target Viewer Function The following table lists available settings items with corresponding displays on the LCD monitor. Furthermore, MEA-Mode combinations allow the measurement of a variety of values, such as maximum and minimum values.
KEYENCE LS-7600 User Manual
Calibration setting error Remedy Check the wiring of the connection cable. The following section provides information on respective command formats. Calibration settings correct for this error. PLC The following environment settings are fixed. The following types of measured value output are possible. If only a single measuring head is used, connect the cable to the HEAD1 connector. Share your thoughts with other customers. Page 82 Chapter 5 Function Settings Change of Edge Detection Threshold The following section provides information on how to make threshold changes for the measurement of highly transparent objects.
Area Selection Area When a target is located in the measuring area as shown below, the light into the receiver is partly obstructed, thus creating a shadow.
1 PC KEYENCE Ls-7601 CCD Laser Micrometer
The backlight does not turn ON. The range of values used for judgment is called a tolerance. The results of difference checks will be output externally, provided that DIFF function output settings are made properly in advance. The minimum value during the sampling period specified is displayed and output. Page Appendix Index 1 One-point calibration The diameter of the circumscribed circle made by the turning object is measured.
A rise in inner temperature may cause equipment failure. Page Chapter 5 Function Settings 5 The result of statistical processing is leyence. ComiXology Thousands of Digital Comics. Page 55 Chapter 4 Operation Control Note: Refer to Timing Chart on page for response delays. The cursor moves to the number of the program to be copied.
Used Keyence LS, LSM for sale by Optical Innovations Inc. |
Don’t see meyence manual you are looking for? Amazon Second Chance Pass it on, trade it in, give it a second life. If the [ENT] key Up or down key is pressed continuously, the set value will automatically increase or decrease. The average value during the sampling period specified is displayed and output.